The widespread use of medical ultrasonic cleaning machines in hospitals is now very wide. Many medical devices or things need to be reused because of raw materials and cost. However, because of the special sanitary conditions of the hospital, these medical devices require medical ultrasonic cleaning machines. Cleaning. The medical ultrasonic cleaning machine sometimes presents problems during the application process. Let's find out what specific common problems?
First, the perforation of the oscillation surface
The oscillating surface in the medical ultrasonic cleaning machine is connected to the transducer to generate large-area ultrasonic waves. When the cleaning machine is in a long-term use, it is easy to overload the oscillating surface and damage the oscillating surface. When it reaches a certain service life, it will The oscillating surface perforation of the washing machine is shown, and the perforation of the stainless steel oscillating surface cannot be reversed and repaired. Generally, the entire machine can only be replaced.
Second, the ultrasonic vibrator degumming
The vibrator of a good quality medical ultrasonic cleaning machine is installed with screws and glue to fix it, so there may be no abnormality in the current when degumming. However, the problem of fixation will cause the vibrator to hit the oscillating surface, which also causes the oscillating surface to be damaged For one reason, then you should pay attention to touching the vibrator to see if it is stable. If you feel loose, you need to do maintenance.
Third, the ultrasonic vibrator is wet
When using a medical ultrasonic cleaning machine , it is necessary to put a cleaning solution. If the solution is not cleaned in time after the cleaning is completed, it is easy to make the ultrasonic vibrator appear wet. In addition, if the surrounding environment of the ultrasonic cleaning machine is humid, This will cause this situation. The damper will affect the current and resistance value when it is wet. The effect of cleaning medical equipment will be greatly reduced. You can remove the shaker for artificial drying, but pay attention to remove the plastic spray casing.
The above are some of the problems and general solutions of medical ultrasonic cleaning machines introduced by Goodyear Ultrasonics for everyone. Although the price of medical ultrasonic cleaning machines is relatively modest now, we must still pay attention to its maintenance and repair to avoid impact. For the work efficiency of a hospital, a professional medical ultrasonic cleaning machine must be selected to ensure the hygiene of medical equipment.